
New TopShim run:

version = topshim 2.1.0
date = 2021-12-22
time = 18:24:51.409
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 4.4 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 2: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 3: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 4: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 5: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.37


Performing tune after shimming
tune lock level increase = 1.1

shim changes: 
Z	+102
Z2	-14
Z3	+28
Z4	+416
Z5	-139
Z6	-2570
Z7	+59
Z8	+12956
X	+17
Y	+62
YZ	+8

duration = 1 min 36 sec
completed successfully
finished Wed Dec 22 18:26:26 2021


========================================================================
