
New TopShim run:

version = topshim 2.0.0
date = 2021-05-15
time = 03:29:10.047
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 2.4 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration  2: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.2 Hz
Iteration  3: FWHM Before= 0.2 Hz, Target= 0.2 Hz, After= 0.8 Hz
Iteration  4: FWHM Before= 0.8 Hz, Target= 0.2 Hz, After= 0.2 Hz
Iteration  5: FWHM Before= 0.2 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.879


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	+56
Z2	+217
Z3	+6
Z4	-5388
Z5	-331

duration = 48 sec
completed successfully
finished Sat May 15 03:29:55 2021


========================================================================
