
New TopShim run:

version = topshim 2.0.0
date = 2021-05-08
time = 05:22:14.301
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 5.0 Hz, Target= 0.6 Hz, After= 2.4 Hz
Iteration  2: FWHM Before= 2.4 Hz, Target= 0.3 Hz, After= 3.6 Hz
Iteration  3: FWHM Before= 3.6 Hz, Target= 0.8 Hz, After= 5.1 Hz
Iteration  4: FWHM Before= 5.1 Hz, Target= 0.2 Hz, After= 3.9 Hz
Iteration  5: FWHM Before= 3.9 Hz, Target= 0.7 Hz, After= 2.9 Hz
Iteration  6: FWHM Before= 2.9 Hz, Target= 0.3 Hz, After= 3.9 Hz
Iteration  7: FWHM Before= 3.9 Hz, Target= 0.8 Hz, After= 2.9 Hz
Iteration  8: FWHM Before= 2.9 Hz, Target= 0.2 Hz, After= 3.1 Hz
Unusually slow convergence detected. Shimming was aborted.
Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, spinning) = 2.25


Sample position (Shigemi) = -0.53 - 0.65 cm.

shim changes: 
Z	+140
Z2	-133
Z3	+460
Z4	+2251
Z5	-5131

duration = 1 min 11 sec
completed successfully
finished Sat May  8 05:23:22 2021


========================================================================
