NMRbot_cq_18971_SampleReport.20150220_1545.txt
Status Report for Automated Sample Data Collection
## Host = kerry.nmrfam.wisc.edu ##

## NMRbot version 20140226 ##

## Initiated - 20150220_1545 ##

- info:
  cq_18971 in DMSO at position 201
  foldername => cq_18971.DMSO.201.20150220
  Exps => ZGCPPR,metabTOCSY,C13CPD,C13DEPT90,C13DEPT135_MMCD,HSQCETGP,HSQCETGP,HMBCETGPL3ND,COSY45SW,HSQC_TOCSY_ADIA
- instrument initialization - passed
- loading sample position 201 - passed
- DMSO solvent lock - passed
-   tuning probe channels - passed
-   topshim tuneb tunea - passed
- creating exp 1 with ZGCPPR parameters
-   prosol parameters loaded
-   NS changed to 24
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 2 with metabTOCSY parameters
-   prosol parameters loaded
-   NS changed to 24
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 3 with C13CPD parameters
-   prosol parameters loaded
-   NS changed to 3072
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 4 with C13DEPT90 parameters
-   prosol parameters loaded
-   NS changed to 768
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 5 with C13DEPT135_MMCD parameters
-   prosol parameters loaded
-   NS changed to 768
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 6 with HSQCETGP parameters
-   prosol parameters loaded
-   NS changed to 6
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 7 with HSQCETGP parameters
-   prosol parameters loaded
-   NS changed to 6
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 8 with HMBCETGPL3ND parameters
-   prosol parameters loaded
-   NS changed to 12
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 9 with COSY45SW parameters
-   prosol parameters loaded
-   NS changed to 24
-   optimizing receiver gain - passed
-   receiver gain set to 203
- creating exp 10 with HSQC_TOCSY_ADIA parameters
-   prosol parameters loaded
-   NS changed to 24
-   optimizing receiver gain - passed
-   receiver gain set to 203
- experiment folders set up - passed
## Data Acquisition ##
# Exp 1 - ZGCPPR (zgcppr)
- dims 1- F1 1H, SW 16.03
- start 20150220_1555
- end   20150220_1557 - passed
-   1H adaptive offset determined = 4.7498
-   1H adaptive SW determined = 11.6991
# Exp 2 - metabTOCSY (mlevph)
- dims 2- F1 1H, SW 10.21; F2 1H, SW 10.21
- start 20150220_1557
- end   20150220_1953 - passed
# Exp 3 - C13CPD (zgpg30)
- dims 1- F1 13C, SW 236.77
- start 20150220_1953
- end   20150220_2235 - passed
# Exp 4 - C13DEPT90 (deptsp90)
- dims 1- F1 13C, SW 236.77
- start 20150220_2235
- end   20150220_2316 - passed
# Exp 5 - C13DEPT135_MMCD (deptsp135)
- dims 1- F1 13C, SW 236.77
- start 20150220_2316
- end   20150220_2356 - passed
-   13C adaptive offset determined = 65.5471
-   13C adaptive SW determined = 139.9272
# Exp 6 - HSQCETGP (hsqcetgp)
- dims 2- F1 13C, SW 165.64; F2 1H, SW 12.99
- start 20150220_2356
- end   20150221_0038 - passed
# Exp 7 - HSQCETGP (hsqcetgp) - SWadapted
-   F1 SW changed to 139.93
-   F2 SW changed to 11.70
-   F1 1H offset changed to 4.75
-   F2 13C offset changed to 65.55
- dims 2- F1 13C, SW 139.93; F2 1H, SW 11.69
- start 20150221_0038
- end   20150221_0120 - passed
# Exp 8 - HMBCETGPL3ND (hmbcetgpl3nd)
- dims 2- F1 13C, SW 221.83; F2 1H, SW 12.02
- start 20150221_0120
- end   20150221_0325 - passed
# Exp 9 - COSY45SW (cosyqf45)
- dims 2- F1 1H, SW 10.31; F2 1H, SW 10.31
- start 20150221_0325
- end   20150221_0518 - passed
# Exp 10 - HSQC_TOCSY_ADIA (hsqcdietgpsisp.2)
- dims 2- F1 13C, SW 165.00; F2 1H, SW 12.02
- start 20150221_0518
- end   20150221_0911 - passed

## cq_18971 Experiment Set Ended - 20150221_0911 ##

