
New TopShim run:

version = topshim 2.1.0
date = 2021-12-23
time = 04:31:42.808
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 4.9 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 2: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 1.3 Hz
Iteration 3: FWHM Before= 1.3 Hz, Target= 0.2 Hz, After= 1.0 Hz
Iteration 4: FWHM Before= 1.0 Hz, Target= 0.2 Hz, After= 0.4 Hz
Iteration 5: FWHM Before= 0.4 Hz, Target= 0.2 Hz, After= 0.8 Hz
Iteration 6: FWHM Before= 0.8 Hz, Target= 0.2 Hz, After= 0.4 Hz
Slow convergence detected. Shimming was aborted.
Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, non-spinning) = 0.545


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	-81
Z2	-40
Z3	+168
Z4	+537
Z5	-711
Z6	-3390
Z7	+3732
Z8	+18205
X	+20
XZ	+92
Y	+1
YZ	+3

duration = 1 min 52 sec
completed successfully
finished Thu Dec 23 04:33:33 2021


========================================================================
