
New TopShim run:

version = topshim 2.1.0
date = 2021-12-23
time = 04:06:46.022
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 2.2 Hz, Target= 0.3 Hz, After= 1.0 Hz
Iteration 2: FWHM Before= 1.0 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 3: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.3 Hz
Iteration 4: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.436


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+138
Z2	-19
Z3	+23
Z4	+526
Z5	-88
Z6	-2849
Z7	-687
Z8	+13043
X	-4
XZ	+8
Y	+12
YZ	-106

duration = 1 min 40 sec
completed successfully
finished Thu Dec 23 04:08:24 2021


========================================================================
