
New TopShim run:

version = topshim 2.1.0
date = 2021-12-22
time = 16:43:51.873
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 5.6 Hz, Target= 0.4 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.551


Performing tune after shimming

NOTE: The lock signal appears to be unstable. This affects the performance of the tune algorithm! 

tune lock level increase = 0.9

shim changes: 
Z	+46
Z2	-10
Z3	+52
Z4	+175
Z5	-257
Z6	-1383
Z7	+1223
Z8	+13022
X	+15
XZ	+10
Y	-1

duration = 1 min 32 sec
completed successfully
finished Wed Dec 22 16:45:22 2021


========================================================================
