
New TopShim run:

version = topshim 2.1.0
date = 2021-12-22
time = 11:17:27.008
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.2 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 2: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.8 Hz
Iteration 3: FWHM Before= 0.8 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration 4: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration 5: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 6: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.8 Hz
Slow convergence detected. Shimming was aborted.
Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, non-spinning) = 0.438


Performing tune after shimming
tune lock level increase = 1.1

shim changes: 
Z	+105
Z2	-1
Z3	+4
Z4	+250
Z5	-65
Z6	-1674
Z7	-555
Z8	+9823
X	+8
XZ	+28
Y	+30
YZ	+7

duration = 1 min 46 sec
completed successfully
finished Wed Dec 22 11:19:11 2021


========================================================================
