
New TopShim run:

version = topshim 2.1.0
date = 2021-12-22
time = 08:31:57.016
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 3.9 Hz, Target= 0.4 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.48


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+94
Z2	-6
Z3	+36
Z4	+239
Z5	-154
Z6	-1630
Z7	-36
Z8	+11952
X	+32
XZ	+30
Y	+33
YZ	-1

duration = 1 min 13 sec
completed successfully
finished Wed Dec 22 08:33:08 2021


========================================================================
