
New TopShim run:

version = topshim 2.0.0
date = 2021-05-15
time = 05:41:02.907
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.9 Hz, Target= 0.3 Hz, After= 3.0 Hz
Iteration  2: FWHM Before= 3.0 Hz, Target= 0.3 Hz, After= 0.3 Hz
Iteration  3: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.421


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	+92
Z2	+67
Z3	+131
Z4	-1148
Z5	-1736

duration = 37 sec
completed successfully
finished Sat May 15 05:41:37 2021


========================================================================
