
New TopShim run:

version = topshim 2.1.0
date = 2021-12-21
time = 05:04:23.624
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 3.9 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 2: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.9 Hz
Iteration 3: FWHM Before= 0.9 Hz, Target= 0.4 Hz, After= 0.2 Hz
Iteration 4: FWHM Before= 0.2 Hz, Target= 0.3 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.486


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+96
Z2	-5
Z3	+44
Z4	+258
Z5	-163
Z6	-2147
Z7	-120
Z8	+12927
X	+8
XZ	-2
Y	+62
YZ	-108

duration = 1 min 56 sec
completed successfully
finished Tue Dec 21 05:06:18 2021


========================================================================
