
New TopShim run:

version = topshim 2.1.0
date = 2021-12-24
time = 00:11:01.947
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 4.0 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration 2: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.3 Hz
Iteration 3: FWHM Before= 0.3 Hz, Target= 0.4 Hz, After= 0.8 Hz
Iteration 4: FWHM Before= 0.8 Hz, Target= 0.4 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.419


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+92
Z2	-17
Z3	+34
Z4	+409
Z5	-172
Z6	-2491
Z7	+89
Z8	+13065
X	+9
XZ	+31
YZ	+8

duration = 1 min 33 sec
completed successfully
finished Fri Dec 24 00:12:33 2021


========================================================================
