
New TopShim run:

version = topshim 2.1.0
date = 2021-12-23
time = 21:37:07.129
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.3 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 2: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.5 Hz
Iteration 3: FWHM Before= 0.5 Hz, Target= 0.3 Hz, After= 0.1 Hz
Iteration 4: FWHM Before= 0.1 Hz, Target= 0.3 Hz, After= 0.3 Hz
Iteration 5: FWHM Before= 0.3 Hz, Target= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, non-spinning) = 0.385


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+59
Z2	-10
Z3	+17
Z4	+340
Z5	-165
Z6	-2307
Z7	+615
Z8	+12892
X	+9
XZ	+9
YZ	-59

duration = 1 min 33 sec
completed successfully
finished Thu Dec 23 21:38:38 2021


========================================================================
