
New TopShim run:

version = topshim 2.1.0
date = 2021-12-23
time = 09:10:44.182
command = topshim tunea

FINE SHIMMING

Parameters:

probe = Z168276_0016
solvent = H2O+DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.70 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 1.7 Hz, Target= 0.4 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.407


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+62
Z2	-31
Z3	+49
Z4	+554
Z5	-277
Z6	-2935
Z7	+1266
Z8	+17500
X	+15
XZ	+16
Y	-3

duration = 1 min 24 sec
completed successfully
finished Thu Dec 23 09:12:06 2021


========================================================================
