
New TopShim run:

version = topshim 2.0.0
date = 2021-05-11
time = 03:35:03.998
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 2.3 Hz, Target= 0.3 Hz, After= 3.9 Hz
Iteration  2: FWHM Before= 3.9 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.503


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	+40
Z2	+90
Z3	+168
Z4	-2091
Z5	-1670

duration = 36 sec
completed successfully
finished Tue May 11 03:35:37 2021


========================================================================
