
New TopShim run:

version = topshim 2.0.0
date = 2021-05-11
time = 03:12:52.496
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 4.1 Hz, Target= 0.4 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.862


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	-206
Z2	+18
Z3	-290
Z4	-1892
Z5	+1844

duration = 31 sec
completed successfully
finished Tue May 11 03:13:20 2021


========================================================================
