
New TopShim run:

version = topshim 2.0.0
date = 2021-05-09
time = 19:53:40.588
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.4 Hz, Target= 0.2 Hz, After= 4.9 Hz
Iteration  2: FWHM Before= 4.9 Hz, Target= 0.2 Hz, After= 0.6 Hz
Iteration  3: FWHM Before= 0.6 Hz, Target= 0.2 Hz, After= 0.8 Hz
Iteration  4: FWHM Before= 0.8 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.741


Sample position (Shigemi) = -0.53 - 0.65 cm.

shim changes: 
Z	+189
Z2	-18
Z3	+567
Z4	-387
Z5	-5143

duration = 41 sec
completed successfully
finished Sun May  9 19:54:18 2021


========================================================================
