
New TopShim run:

version = topshim 2.0.0
date = 2021-05-09
time = 12:14:58.965
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 0.2 Hz, Target= 0.3 Hz, After= 7.3 Hz
Iteration  2: FWHM Before= 7.3 Hz, Target= 0.3 Hz, After= 6.8 Hz
Iteration  3: FWHM Before= 6.8 Hz, Target= 0.6 Hz, After= 0.3 Hz
Iteration  4: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 9.6 Hz
Iteration  5: FWHM Before= 9.6 Hz, Target= 0.5 Hz, After=10.3 Hz
Iteration  6: FWHM Before=10.3 Hz, Target= 0.6 Hz, After= 1.9 Hz
Iteration  7: FWHM Before= 1.9 Hz, Target= 0.6 Hz, After= 0.6 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.847


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	-153
Z2	-38
Z3	-220
Z4	+355
Z5	+1652

duration = 1 min 3 sec
completed successfully
finished Sun May  9 12:15:59 2021


========================================================================
