
New TopShim run:

version = topshim 2.0.0
date = 2021-05-07
time = 01:02:27.296
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 4.1 Hz, Target= 0.2 Hz, After= 8.4 Hz
Iteration  2: FWHM Before= 8.4 Hz, Target= 0.7 Hz, After= 8.4 Hz
Iteration  3: FWHM Before= 8.4 Hz, Target= 0.3 Hz, After= 1.2 Hz
Iteration  4: FWHM Before= 1.2 Hz, Target= 0.7 Hz, After= 0.8 Hz
Iteration  5: FWHM Before= 0.8 Hz, Target= 0.2 Hz, After= 9.0 Hz
Iteration  6: FWHM Before= 9.0 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.816


Sample position (Shigemi) = -0.53 - 0.65 cm.

shim changes: 
Z	-97
Z2	+261
Z3	+251
Z4	-7183
Z5	-802

duration = 54 sec
completed successfully
finished Fri May  7 01:03:18 2021


========================================================================
