
New TopShim run:

version = topshim 2.0.0
date = 2021-05-10
time = 13:45:56.829
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 4.6 Hz, Target= 0.2 Hz, After= 1.3 Hz
Iteration  2: FWHM Before= 1.3 Hz, Target= 0.3 Hz, After= 1.2 Hz
Iteration  3: FWHM Before= 1.2 Hz, Target= 0.7 Hz, After= 0.7 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.863


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	-248
Z2	-61
Z3	-478
Z4	-1085
Z5	+1750

duration = 39 sec
completed successfully
finished Mon May 10 13:46:33 2021


========================================================================
