
New TopShim run:

version = topshim 2.0.0
date = 2021-05-10
time = 11:47:08.079
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 4.0 Hz, Target= 0.5 Hz, After= 2.5 Hz
Iteration  2: FWHM Before= 2.5 Hz, Target= 0.4 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.58


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	+32
Z2	+34
Z3	+113
Z4	-649
Z5	-2030

duration = 33 sec
completed successfully
finished Mon May 10 11:47:38 2021


========================================================================
