
New TopShim run:

version = topshim 2.0.0
date = 2021-05-10
time = 07:11:52.969
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 2.4 Hz, Target= 0.5 Hz, After= 1.8 Hz
Iteration  2: FWHM Before= 1.8 Hz, Target= 0.5 Hz, After= 1.3 Hz
Iteration  3: FWHM Before= 1.3 Hz, Target= 0.5 Hz, After= 0.9 Hz
Iteration  4: FWHM Before= 0.9 Hz, Target= 0.5 Hz, After= 1.4 Hz
Iteration  5: FWHM Before= 1.4 Hz, Target= 0.4 Hz, After= 1.1 Hz
Iteration  6: FWHM Before= 1.1 Hz, Target= 0.5 Hz, After= 0.6 Hz
Iteration  7: FWHM Before= 0.6 Hz, Target= 0.5 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.598


Sample position (Shigemi) = -0.53 - 0.66 cm.

shim changes: 
Z	+49
Z2	+32
Z3	+64
Z4	-903
Z5	-1677

duration = 58 sec
completed successfully
finished Mon May 10 07:12:48 2021


========================================================================
