
New TopShim run:

version = topshim 2.0.0
date = 2021-05-10
time = 03:31:17.334
command = topshim  fine h2o_1.7mm ordmax=5 shigemi

FINE SHIMMING

Parameters:

probe = Z150329_0015
solvent = 5%DMSOinH2O_1.7mm
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.59 ppm

Results:


1D Shimming:
Iteration  1: FWHM Before= 2.8 Hz, Target= 0.3 Hz, After= 3.0 Hz
Iteration  2: FWHM Before= 3.0 Hz, Target= 0.2 Hz, After= 0.9 Hz
Iteration  3: FWHM Before= 0.9 Hz, Target= 0.4 Hz, After= 5.7 Hz
Iteration  4: FWHM Before= 5.7 Hz, Target= 0.3 Hz, After= 4.8 Hz
Iteration  5: FWHM Before= 4.8 Hz, Target= 0.2 Hz, After= 0.9 Hz
Iteration  6: FWHM Before= 0.9 Hz, Target= 0.5 Hz, After= 6.1 Hz
Iteration  7: FWHM Before= 6.1 Hz, Target= 0.3 Hz, After= 5.7 Hz
Iteration  8: FWHM Before= 5.7 Hz, Target= 0.3 Hz, After= 0.8 Hz
Iteration  9: FWHM Before= 0.8 Hz, Target= 0.3 Hz, After= 1.0 Hz
Iteration 10: FWHM Before= 1.0 Hz, Target= 0.4 Hz, After= 5.5 Hz
Iteration 11: FWHM Before= 5.5 Hz, Target= 0.3 Hz, After= 4.4 Hz
Iteration 12: FWHM Before= 4.4 Hz, Target= 0.2 Hz, After= 0.7 Hz
Iteration 13: FWHM Before= 0.7 Hz, Target= 0.3 Hz, After= 2.7 Hz
Iteration 14: FWHM Before= 2.7 Hz, Target= 0.3 Hz, After= 6.7 Hz
Iteration 15: FWHM Before= 6.7 Hz, Target= 0.3 Hz, After= 4.5 Hz
RMSD between target and map above threshold but maximum number of iterations reached. Halting shimming.

Using the best result as the final shim state.

Lineshape quality factor (1D spectrum, spinning) = 1.25


Sample position (Shigemi) = -0.53 - 0.65 cm.

shim changes: 
Z	-92
Z2	-78
Z3	+224
Z4	+2112
Z5	-673

duration = 1 min 45 sec
completed successfully
finished Mon May 10 03:32:59 2021


========================================================================
